The heat exchanger on a Toshiba Aquilion CT plays a crucial role in the overall function of the scanner. Just like the radiator and engine in your car, the heat exchanger keeps the x-ray tube cool. If defective, then the system is unable to perform scans–the engine, so to speak, won’t run.
So how do you know if your heat exchanger is defective?
Today we’re discussing how to diagnose a defective heat exchanger on a Toshiba/Canon Aquilion CT system. Image 1 displays a common error message, which is usually caused by a defective heat exchanger.
HOW TO DIAGNOSE A DEFECTIVE HEAT EXCHANGER:
To see if the heat exchanger is causing the error, follow these steps:
Step One: Check to see if there is power applied to the heat exchanger.
Step Two: Make sure the heat exchanger fan filter is not clogged and all four fans are spinning.
Step Three: If the generator type is SRU:
- Place a jumper wire between pins 7 and 8 of TB5 on the SRU power unit. If the error clears, the heat exchanger is defective.
Step Four: If the generator type is Spellman:
- Place a jumper wire between pins 7 and 8 of TB1 on the inverter unit. If the error clears, the heat exchanger is defective.
As a leader in the Toshiba Aquilion CT market, we are proud to offer Heat Exchangers that are completely rebuilt to zero-hour specs—at less than one fourth the OEM cost. Richardson Healthcare can help you provide the best CT service possible. This is a small preview of our Toshiba/Canon Aquilion training course covering the 4, 16, 32/64, and Prime scanners. Ask us for details!
All product and company names are trademarks™ or registered® trademarks of their respective holders. Use of the trademarks is solely for identification purposes, and does not imply any affiliation with or endorsement by the trademark holders.